This is the second edition of CAN/CSA-ISO 13958, Concentrates for haemodialysis and related therapies, which is an adoption without modification of the identically titled ISO (International Organization for Standardization) Standard 13958 (third edition, 2014-04-01). It supersedes the previous edition published in 2011 as CAN/CSA-ISO 13958 (adopted ISO 13958:2009). At the time of publication, ISO 13958:2014 is available from ISO in English only. CSA Group will publish the French version when it becomes available from ISO.
This International Standard specifies minimum requirements for concentrates used for haemodialysis and related therapies. For the purpose of this International Standard, concentrates are a mixture of chemicals and water, or chemicals in the form of dry powder or other highly concentrated media, that are delivered to the end user to make dialysis fluid used to perform haemodialysis and related therapies. This International Standard is addressed to the manufacturer of such concentrates. In several instances in this International Standard, it became necessary to address the dialysis fluid, which is made by the end user, to help clarify the requirements for manufacturing concentrates. Because the manufacturer of the concentrate does not have control over the final dialysis fluid, any reference to dialysis fluid is for clarification and is not a requirement of the manufacturer.
This International Standard includes concentrates in both liquid and powder forms. Also included are additives, also called spikes, which are chemicals that may be added to the concentrate to increase the concentration of one or more of the existing ions in the concentrate and thus in the final dialysis fluid. This International Standard also gives requirements for equipment used to mix acid and bicarbonate powders into concentrate at the user’s facility.
Concentrates prepared from prepackaged salts and water at a dialysis facility for use in that facility are excluded from the scope of this International Standard. Although references to dialysis fluid appear herein, this International Standard does not address dialysis fluid as made by the end user. Also excluded from the scope of this International Standard are requirements for the monitoring frequency of water purity used for the making of dialysis fluid by the dialysis facility. Recommendations from the technical committee responsible for this International Standard for monitoring water quality are contained in ISO 23500. This International Standard does not address bags of sterile dialysis fluid or sorbent dialysis fluid regeneration systems that regenerate and recirculate small volumes of the dialysis fluid.
Concentrates for haemodialysis and related therapies (Adopted ISO 13958:2009, second edition, 2009-04-15)