Codes & Standards - Purchase
The choice of method depends on the equipment available, the matrix and the concentration range of interest. Both methods are suitable for the determination of mercury in water. The method with enrichment commonly has a practical working range from 0,01 µg/l to 1 µg/l. The mean limit of quantification (LOQ) reported by the participants of the validation trial was 0,008 µg/l. This information on the LOQ gives the user of ISO 12846:2012 an orientation and does not replace the estimation of performance data from laboratory specific data. It has to be considered that it is possible to achieve lower LOQs with specific instrumentation (e.g. single mercury analysers).
The method without enrichment commonly has a practical working range starting at 0,05 µg/l. The LOQ reported by the participants of the validation trial was 0,024 µg/l. It is up to the user, based on the specific application, to decide whether higher concentrations are determined by omitting the enrichment step and/or by diluting the sample(s). The sensitivity of both methods is dependent on the selected operating conditions.
Another possibility for the determination of extremely low Hg concentrations down to 0,002 µg/l without pre-concentration is the application of atomic fluorescence spectrometry (see ISO 17852). Specific atomic-absorption mercury analysers allow determinations down to 0,010 µg/l without pre-concentration.
In general, the determination of trace concentrations of Hg by AAS (or AFS) is dependent on clean operating conditions in the laboratory and on the use of high-purity chemicals with negligible low-Hg blanks.
Note that ISO 12846:2012 may be applied to industrial and municipal waste water after an additional digestion step performed under appropriate conditions and after suitable method validation. A potential sample stability issue (mercury loss) for anaerobic reducing industrial effluents has to be considered thoroughly.