Codes & Standards - Purchase
CSA ISO 11737-2:21
Sterilization of health care products — Microbiological methods — Part 2: Tests of sterility performed in the definition, validation and maintenance of a sterilization process (Adopted ISO 11737-2:2019, third edition, 2019-12)
View Access for this document is only available for viewers in Canada
This is the first edition of CSA ISO 11737-2, Sterilization of health care products — Microbiological methods — Part 2: Tests of sterility performed in the definition, validation and maintenance of a sterilization process, which is an adoption without modification of the identically titled ISO (International Organization for Standardization) Standard 11737-2 (third edition, 2019-12).
CSA Group acknowledges that the adoption of this Standard was made possible, in part, by the financial support of the governments of Alberta, British Columbia, Manitoba, New Brunswick, Newfoundland and Labrador, Northwest Territories, Nova Scotia, Nunavut, Ontario, Prince Edward Island, Quebec, Saskatchewan, and Yukon, as administered by the Canadian Agency for Drugs and Technologies in Health (CADTH).
This Standard has been developed in compliance with Standards Council of Canada requirements for National Standards of Canada. It has been published as a National Standard of Canada by CSA Group.
This document specifies the general criteria for tests of sterility on medical devices that have been exposed to a treatment with the sterilizing agent which has been reduced relative to that anticipated to be used in routine sterilization processing. These tests are intended to be performed when defining, validating or maintaining a sterilization process.
This document is not applicable to:
a) sterility testing for routine release of product that has been subjected to a sterilization process,
b) performing a test for sterility (see 3.12),
NOTE 1 The performance of a) or b) is not a requirement of ISO 11135, ISO 11137-1, ISO 11137-2, ISO 14160, ISO 14937, ISO 17665-1 or ISO 20857.
c) test of sterility or test for sterility for demonstration of product shelf life, stability and/or package integrity, and
d) culturing of biological indicators or inoculated products.
NOTE 2 Guidance on culturing biological indicators is included in ISO 11138-7.