This is the first edition of CAN/CSA-C22.2 No. 80601-2-67, Medical electrical equipment - Part 2-67: Particular requirements for basic safety and essential performance of oxygen conserving equipment, which is an adoption, with Canadian deviations, of the identically titled ISO (International Organization for Standardization) Standard 80601-2-67 (first edition, 2014-06-01). It replaces CAN/CSA-Z18779:08, Medical devices for conserving oxygen and oxygen mixtures - Particular requirements (adopted ISO 18779:2005). It is one in a series of Standards issued by CSA Group under Part II of the Canadian Electrical Code.
For brevity, this Standard will be referred to as CAN/CSA-C22.2 No. 80601-2-67 throughout.
This Standard is intended to be used in conjunction with CAN/CSA-C22.2 No. 60601-1:14, Medical electrical equipment - Part 1: General requirements for basic safety and essential performance (adopted IEC 60601-1:2005, including Amendment 1:2012, with Canadian deviations) and CAN/CSAC22.2 No. 60601-1-11:15, Medical electrical equipment - Part 1-11: General requirements for basic safety and essential performance - Collateral standard: Requirements for medical electrical equipment and medical electrical systems used in the home healthcare environment (adopted IEC 60601-1-11:2015, with Canadian deviations).
Scope and object
IEC 60601-1:2005+A1:2012, Clause 1 applies, except as follows:
201.1.1 * Scope
IEC 60601-1:2005+A1:2012, 1.1 is replaced by:
This particular standard is applicable to the BASIC SAFETY and ESSENTIAL PERFORMANCE of oxygen CONSERVING EQUIPMENT, hereafter referred to as ME EQUIPMENT, in combination with its ACCESSORIES intended to conserve supplemental oxygen by delivering gas intermittently and synchronized with the PATIENT'S inspiratory cycle, when used in the HOME HEALTHCARE ENVIRONMENT. Oxygen CONSERVING EQUIPMENT is typically used by a LAY OPERATOR.
NOTE 1 CONSERVING EQUIPMENT can also be used in professional health care facilities.
NOTE 2 CONSERVING EQUIPMENT can be used with an oxygen concentrator.
This particular standard is also applicable to those ACCESSORIES intended by their MANUFACTURER to be connected to CONSERVING EQUIPMENT, where the characteristics of those ACCESSORIES can affect the BASIC SAFETY or ESSENTIAL PERFORMANCE of the CONSERVING EQUIPMENT.
This particular standard is only applicable to active devices (e.g. pneumatically or electrically powered) and is not applicable to non-active devices (e.g. reservoir cannulas).
NOTE 3 CONSERVING EQUIPMENT complying with this particular standard can be incorporated with other devices that have their own standards, in which case the combination needs to comply with both standards.
EXAMPLES CONSERVING EQUIPMENT combined with a pressure regulator , an oxygen concentrator  or liquid oxygen equipment .
If a clause or subclause is specifically intended to be applicable to ME EQUIPMENT only, or to ME SYSTEMS only, the title and content of that clause or subclause will say so. If that is not the case, the clause or subclause applies both to ME EQUIPMENT and to ME SYSTEMS, as relevant.
HAZARDS inherent in the intended function of ME EQUIPMENT or ME SYSTEMS within the scope of this standard are not covered by specific requirements in this standard except in 7.2.13 and 8.4.1 of the general standard.
NOTE 4 See also 4.2 of the general standard.
This particular standard is a particular standard in the IEC 60601 series of standards.
IEC 60601-1:2005, 1.2 is replaced by:
The object of this particular standard is to establish particular BASIC SAFETY and ESSENTIAL PERFORMANCE requirements for CONSERVING EQUIPMENT [as defined in 201.3.201] and its ACCESSORIES.
NOTE ACCESSORIES are included because ACCESSORIES can have a significant impact on the BASIC SAFETY or ESSENTIAL PERFORMANCE of CONSERVING EQUIPMENT.
items 1 - of 4 result(s)
View 20 | View 100
Medical electrical equipment — Part 2-67: Particular requirements for basic safety and essential performance of oxygen conserving equipment (Adopted ISO 80601-2-67:2014, first edition, 2014-06-01, with Canadian deviations)
Appareils électromédicaux — Partie 2-67 : Exigences particulières pour la sécurité de base et les performances essentielles des économiseurs d’oxygène (norme ISO 80601-2-67:2014 adoptée, première édition, 2014-06-01, avec exigences propres au Canada)
Medical Devices for Conserving Oxygen and Oxygen Mixtures - Particular Requirements (Adopted ISO 18779:2005, first edition, 2005-02-15, with Canadian deviations)
Économiseurs médicaux d'oxygène et de mélanges oxygénés - Exigences particulières (norme ISO 18779:2005 adoptée, première édition, 2005-02-15, avec exigences propres au Canada)